Central Tracking for experiment BM&N based on double side silicon detectors.
Double side silicon micro-strip detector invented, designed and assembled for BM@N experiment is considered. Each plane of detector with common square 250x250mm consists of eight modules. Each module is a 1280ch. 300µm double side DC detector with 90µm pitch for P+ and 103 µm pitch and 2.5° for N+ side. Strips ate bonded to the front-end electronics based on VATAGA7.1 ASICs from IDEAS company. VATAGP7.1 ASIC is a 128ch multiplexed analogue readout and calibration facilities chip . In addition, each channel has a fast shaper that gives a trigger signal.